Precise Controlled Atmosphere Glove Boxes provide a leak-tight environment for work with contamination-sensitive materials. Procedures where low levels of oxygen or moisture are important, such as work with organometallics, alternative energy cells or hydrophilic chemicals, are typical applications for these economical glove boxes.
Need an ultrapure inert atmosphere? All Precise Controlled Atmosphere Glove Boxes are compatible with the new AtmosPure Re-Gen Gas Purifier accessory, which has the capacity to remove up to five liters of oxygen (13.500 ppm) and 660 grams of moisture (18 million ppm) at standard temperature and pressure (STP).
One-piece molded, medium-density polyethylene shell. Labconco recommends this website for a comprehensive listing of chemical resistance data for polyethylene.
Dry powder epoxy-coated steel superstructure
1/4" thick laminated safety glass viewing window, 26.4" wide x 12.5" high, removable for loading equipment, with powder-coated steel frame and neoprene gasket
Space-saving inner and outer transfer chamber doors that pivot upward, are counterbalanced and equipped with quick-latches
Black, chemical-resistant work surface sealed to the glove box floor
Pressure relief valve that activates at pressure/vacuum above 6 inches or below -6 inches of water gauge.
Pressure relief valve vent, 4" nominal diameter
Interior ceiling-mounted electrical duplex receptacle with 5 amp maximum circuit load
Neoprene gloves, size 9 3/4, are included.
Fluorescent lamp, 22 watts, with easy replacement from the top
Switches for light and innermost interior electrical receptacle
Approximately 13 cubic feet of interior space in main chamber
11.0" nominal ID x 12.0" long transfer chamber
Factory leak tested with a mass spectrometer while glove box is pressurized with helium at 5 inches of water gauge.
No detectable leaks greater than 1 x 10(-4) ml/sec
Meet Class 1 atmosphere containment conditions for oxygen leak rates per ISO 10648-2 test method
Six manual valves for purging and filling of both main chamber and transfer chamber and for drying train connections
Main chamber pressure gauge with readings from -5 to +5 inches of water gauge
Transfer chamber pressure gauge with readings from 0 to -29 inches of mercury
Transfer chamber high vacuum sleeve and interior interlocking shelf of dry powder epoxy-coated steel
ETL listed (115 volt models)
CE Conformity (230 volt models)
International electrical configurations
Rotary Vane Direct Drive Vacuum Pumps
Adjustable Height Base Stands
Inert gas for filling
Acrylic Viewing Window
Automatic Pressure Controller
Electrical Power Strips
Exterior Transfer Chamber Shelf
External Glove Port Covers
Gas Fill/Vacuum Purge Tubing Kits
Gloves and Glove Box Accessories
Hanging Shelf Accessories
Oxygen and Moisture Removal Column
Pressure Relief Valve Vent Kits
Storage Shelf Kits
Connecting Gas and Vacuum to a Glove Box
This instructional video shows how to connect gas and vacuum to a Labconco Glove Box with Auto Pressure Controller. Contact Labconco with additional questions not addressed in this video.
Installation Qualification/Operation Qualification for Precise Controlled Atmosphere Glove Box
For validation of predetermined Precise Controlled Atmosphere Glove Box specifications. Provides a checklist of installation and operation qualifications to complete when beginning operation on new or used Precise Controlled Atmosphere Glove Box.