These specialized Class II biological safety cabinets provide user, animal and environmental protection during small animal activities.
PuriCare Procedure Stations provide personnel, animal and environmental protection. They are specialized Class II, Type A2 biological safety cabinets that meet or exceed the product and personnel protection requirements of NSF 49. These Stations are designed for conducting small animal research activities including cage changing, weighing, anesthesia delivery and surgical procedures.
NSF Listed, Standard 49
UL Standard 61010-1 (115 volt models)
CAN/CSA C22.2 No. 1010.1-92 (115 volt models)
CE conformity marking (230 volt models)
Nominal inflow velocity of 105 feet per minute (fpm) with 10" sash opening; 85 fpm with 12" sash opening.
Nominal downflow velocity of 55 fpm
Approximately 70% air recirculation
Interior-mounted, line-of-sight LCD information center with "Filter Life Remaining" bar graph, status line for alarm conditions and alerts to warn when filter life diminishes to 20%, 10% and 0%
Filter monitoring system and electronically commutated motor (ECM) that automatically adjusts as filters load
Built-in interval or elapsed timer for experiment monitoring, fluorescent light or UV light control (on models with UV light)
Touchpad control on right-hand side post for manual activation of blower, light, timer, audible alarm mute and menu selection
Radiused type 304 stainless steel interior and removable, seamless, dished work surface with lift out knobs
Powder-coated steel exterior
Towel catch with sponge-type prefilter located under work surface
Contain-Air™ Negative Pressure Channel
Class 5 conditions per ISO 14644-1 and 2 (formerly Class 100)
Supply and exhaust 99.99% efficient HEPA filters. Contact Labconco for optional 99.999% efficient ULPA filters.
Trace odor carbon postfilter
Two electrical duplex receptacles (single outlets on 230 volt models), located one on each side, with ground fault interruption (115 volt models) and stainless steel splash covers
Fully-closing, clear 1/4" tempered safety glass sash with two sash handles; counterbalanced, anti-racking mechanism; and 10° slope
21.7" sash opening and 27.5" viewing height
Curved stainless steel inlet grille with Reserve-Air™ Secondary Airflow Slots
Bright, 90-100 footcandle, glare-free fluorescent lighting located outside the contaminated work area
Intrinsically-safe negative pressure design
Electronic security lock (optional activation) that requires code to operate the cabinet
Full five year warranty on parts and service
Available in 4' and 6' widths
International electrical configurations
254 UV lamp with timer
Chrome-plated forged brass service fixture(s) with ball valve and quarter turn handle, factory installed 8.5" above work surface
Activated Carbon Filters to remove odors
Canopy Connection Kits to allow the PuriCare Station to be thimble-ducted to the outside
PuriCare Laboratory Animal Research Stations Catalog
Contains an overview, features and benefits, ordering information, accessory information and dimensional drawings for PuriCare Open Access Stations, Vertical Flow Stations, Dual Access Laminar Flow Cabinets, Procedure Stations and Bedding Disposal Stations.
Installation Qualification/Operation Qualification for PuriCare Procedure Stations
For validation of predetermined PuriCare Procedure Station specifications. Provides a checklist of installation and operation qualifications to complete when beginning operation on new or used PuriCare Procedure Stations.