Purifier Logic Class II, Type A2 Biosafety Cabinets
Purifier Logic Class II, Type A2 Biosafety Cabinets have been replaced by newer Logic+ models.
They provide personnel, product and environmental protection from hazardous particulates such as agents that require Biosafety Level 1, 2 or 3 containment. Other appropriate applications include work with antineoplastic drugs, genetic material, asbestos and additional substances that generate hazardous airborne particulates. All models are NSF-Listed.
These biological safety cabinets are designed to discharge HEPA-filtered exhaust air directly into the laboratory, or into an exhaust system through the optional Canopy Connection. When canopy-connected, the biosafety cabinets may be used for applications involving minute quantities of volatile toxic chemicals and tracer amounts of radionuclides as an adjunct to microbiological research.
During operation, room air is drawn into the inlet grill. The air in the plenum beneath the work surface is a mixture of unfiltered room air, and air that has just passed through the work area. This contaminated air is drawn by the blower through the back plenum of the cabinet, where approximately 70% of the air is recirculated through the supply HEPA filter and back ove rthe work area. The balance of the contaminated air is discharged to the environment after passing through the exhaust HEPA filter.
View replacement: Purifier Logic+ Class II, Type A2 Biosafety Cabinets (MP-Purifier-Logic+-Class-II,-Type-A2-Biosafety-Cabinets)
- This product has been discontinued and is no longer available for purchase.
- Support Literature is still available for download. Replacement parts and accessories may be available for purchase.
- Please contact Labconco at 800-821-5525 or 816-333-8811
• NSF International Standard 49 for Biohazard Cabinetry
• Nominal inflow velocity of 105 fpm (0.5 m/sec)
• 4’ and 6’ nominal widths
• Remote Blowers
Installing the Purifier Logic Prefilter Kit
Video outlining the process of installing the prefilter kit on Labconco's Purifier Logic Class II Biological Safety Cabinet.
Biosafety Cabinet Filter Testing
How to test filters on Labconco biosafety enclosures.
Labconco Biosafety Cabinet and Glove Box Smoke Test
Labconco Biosafety Cabinet Smoke Split Test
Purifier Logic Biosafety Cabinet Video
Altitude Adjustment for Purifier Logic Biosafety Cabinets
Answers to frequently asked questions in regard to altitude adjustment for Puriﬁer Logic Biosafety Cabinets.
Navigating the Logic Certifier's Menu - Certifier InstructionsLogin to access file!
Navigating the Logic Menu - User InstructionsLogin to access file!
Purifier Logic Bag-In Bag-Out Assembly Instructions
Purifier Logic Class II Type A2 Biosafety Cabinet Manual
For catalog numbers: 34300 Series, 34308 Series, 34400 Series, 34500 Series, 34408 Series, 34508 Series, 34600 Series, 34608 Series
Purifier Logic Unpacking Instructions
Wiring Diagram for the Purifier Logic Class II A2/B2 Biosafety Cabinet, 115 V 50/60Hz operation
Wiring Diagram for the Purifier Logic Class II A2/B2 Biosafety Cabinet, 230V 50/60Hz operation
Logic and PuriCare Technical Manual and Specifications (Before 2013)
Provides detailed information for use in certifying or servicing Labconco biological safety cabinets.
Installation Qualification/Operation Qualification
Installation Qualification/Operation Qualification for Purifier Logic Biosafety Cabinets
Declarations of Conformity
ETL Authorization to Mark Purifier Logic and Cell Logic Biosafety Cabinets
This document from Intertek authorizes the use of the ETL mark on Purifier Logic and Cell Logic Class II Biosafety Cabinets.
CAG-007-2010 Application Guide for Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Controlling Airflow in Class II Biosafety Cabinets
The Advantages of Using an Electronically Commutated Motor (ECM) in Biosafety Cabinets
This white paper explains the ECM motor in our Purifier Logic Biological Safety Cabinets and its advantages over PSC motors.